Browse to https://mendix.princeton.edu/
Select “Request Access” on the login page and fill out the form. The MNFL staff will then authorize you to use the software.&nb
The Tystar Polysilicon Tube (#2) is down due to excessive gasket leakage. The gaskets around the silane injection ring appears to have aged and hardened so that an adequate seal is not made to the quartz tube surface. The tube will have to be cooled to room temperature, the door disassembled, all the gaskets replaced, the door reassembled, the tube leak checked, and the tube reheated. This is tentatively scheduled for Wednesday and Thursday, May 25th and 26th.