
related topics 
{math, energy, light} 
{system, computer, user} 
{line, north, south} 

An ohmmeter is an electrical instrument that measures electrical resistance, the opposition to an electric current. Microohmmeters (microhmmeter or microohmmeter) make low resistance measurements. Megohmmeters (aka megaohmmeter or in the case of a trademarked device Megger) measure large values of resistance. The unit of measurement for resistance is ohms (Ω).
The original design of an ohmmeter provided a small battery to apply a voltage to a resistance. It uses a galvanometer to measure the electric current through the resistance. The scale of the galvanometer was marked in ohms, because the fixed voltage from the battery assured that as resistance is decreased, the current through the meter would increase.
A more accurate type of ohmmeter has an electronic circuit that passes a constant current (I) through the resistance, and another circuit that measures the voltage (V) across the resistance. According to the following equation, derived from Ohm's Law, the value of the resistance (R) is given by:
For highprecision measurements the above types of meter are inadequate. This is because the meter's reading is the sum of the resistance of the measuring leads, the contact resistances and the resistance being measured. To reduce this effect, a precision ohmmeter has four terminals, called Kelvin contacts. Two terminals carry the current from the meter, while the other two allow the meter to measure the voltage across the resistor. With this type of meter, any voltage drop due to the resistance of the first pair of leads and their contact resistances is ignored by the meter. This four terminal measurement technique is called Kelvin sensing, after William Thomson, Lord Kelvin, who invented the Kelvin bridge in 1861 to measure very low resistances. The Fourterminal sensing method can also be utilized to conduct accurate measurements of low resistances.
See also
External links
Full article ▸


related documents 
Superluminal communication 
Antenna effective area 
Spheroid 
North Star 
Angular acceleration 
Frustum 
Gegenschein 
Nova 
Electrooptic effect 
Atomic, molecular, and optical physics 
Fomalhaut 
Iapetus (moon) 
Dynamic mechanical spectroscopy 
Transmission medium 
Albireo 
Rheology 
Maunder Minimum 
Orrery 
Linear polarization 
Pascal (pressure) 
Wide angle Xray scattering 
Miranda (moon) 
Strouhal number 
Ring current 
Electrical length 
Cutback technique 
Gouraud shading 
3753 Cruithne 
Weyl's postulate 
Positron 
