# Time-domain reflectometer

 related topics {system, computer, user} {math, energy, light} {island, water, area} {@card@, make, design} {line, north, south} {food, make, wine}

A time-domain reflectometer (TDR) is an electronic instrument used to characterize and locate faults in metallic cables (for example, twisted wire pairs, coaxial cables).[1] It can also be used to locate discontinuities in a connector, printed circuit board, or any other electrical path. The equivalent device for optical fiber is an optical time-domain reflectometer.

## Contents

### Description

A TDR transmits a short rise time pulse along the conductor. If the conductor is of a uniform impedance and properly terminated, the entire transmitted pulse will be absorbed in the far-end termination and no signal will be reflected toward the TDR. Any impedance discontinuities will cause some of the incident signal to be sent back towards the source. This is similar in principle to radar.

Increases in the impedance create a reflection that reinforces the original pulse whilst decreases in the impedance create a reflection that opposes the original pulse.

The resulting reflected pulse that is measured at the output/input to the TDR is displayed or plotted as a function of time and, because the speed of signal propagation is almost constant for a given transmission medium, can be read as a function of cable length.

Because of this sensitivity to impedance variations, a TDR may be used to verify cable impedance characteristics, splice and connector locations and associated losses, and estimate cable lengths.

### Explanation

Consider the case where the far end of the cable is shorted (that is, it is terminated into zero ohms impedance). When the rising edge of the pulse is launched down the cable, the voltage at the launching point "steps up" to a given value instantly and the pulse begins propagating down the cable towards the short. When the pulse hits the short, no energy is absorbed at the far end. Instead, an opposing pulse reflects back from the short towards the launching end. It is only when this opposing reflection finally reaches the launch point that the voltage at this launching point abruptly drops back to zero, signaling the fact that there is a short at the end of the cable. That is, the TDR had no indication that there is a short at the end of the cable until its emitted pulse can travel down the cable at roughly the speed of light and the echo can return back up the cable at the same speed. It is only after this round-trip delay that the short can be perceived by the TDR. Assuming that one knows the signal propagation speed in the particular cable-under-test, then in this way, the distance to the short can be measured.